New generation of highly sensitive radon detectors based on activated carbon with compensated temperature dependence

Recently patented compensation module for sensors of radioactive noble gases paves the way for novel designs of radon detectors/monitors with adsorbents, suitable for long-term measurements. The module compensates the strong influence of the temperature on the response of detectors with activated carbon or other ad/absorbents. This report describes radon detectors in which activated carbon fabric is coupled with a solid state nuclear track detector and placed in a compensation module. The module is a hermetic volume made of polyethylene foil, through which the radon diffuses. Two reciprocal trends make possible the temperature compensation: that of the radon penetration through the walls, which increases with the increase of the temperature, and that of the adsorption, which decreases. The results show that outside the module the variation of the detector response over the range of 5–35 °C is 230–305%. In contrast, inside the module the variation is reduced to 6–17%. The module also protects the sensor from humidity and thoron, keeping its sensitivity to radon 7–9 times higher than that of commonly used radon detectors. This makes the new detectors very useful for measurement of low radon concentrations in the atmosphere.


Scientific Reports
| (2022) 12:8479 | https://doi.org/10.1038/s41598-022-12502-z www.nature.com/scientificreports/ which leads to substantial loss of adsorption ability, when the occlusion of the pores by water happens 15 . So far the activated carbon-based methods are used mostly for short-term exposures-rarely for more than few days, and usually the activity is measured after the exposure, e. g. by liquid scintillation counting 9 . A novel design of detector modules with compensated temperature dependence has been recently patented 16 . It employs placing detectors which sensitivity decreases with the increase of the temperature, like these based on activated carbon, in hermetic volumes, one or more walls of which are made of plastic material through which radon may diffuse. The fraction of radon penetrated by diffusion inside the volume increases with the increase of the temperature [17][18][19] . The key concept is to design proper "compensation modules", in which these two independent and reciprocal trends compensate each other. In this work detector modules with radon detectors made by activated carbon fabrics coupled with SSNTDs are studied theoretically and experimentally. By computer modeling it is shown that compensation modules with sufficiently compensated temperature dependence might be designed. Experimental tests showed that the variation in the response to radon over 5-35 °C range can be reduced from 230-305% to 6-17%. In the same time the studied detector modules provide 7-9 times better sensitivity than conventional passive radon detectors. The compensation modules may be made thin, flexible and they could be useful for exposure under various environmental conditions. Such modules retard water penetration to the sensor inside, and protect it from thoron ( 220 Rn, half-life 55.6 s) interference on the radon signal. These qualities of the module make possible to build new generation of radon detectors and monitors of much enhanced sensitivity and usable for long-term exposures at highly variable conditions that are typical for the atmospheric environment.

Materials and methods
Concept. The design of the detector module with temperature compensation is shown in Fig. 1. The sensor inside uses an adsorber of activated carbon fabrics coupled with SSNTD of Kodak-Pathe LR-115/II. In the present experiments three kinds of activated carbon fabrics (produced by Kynol Europa GmbH, Germany) were used: Kynol 507-10, ACC-10 and ACC-20. The activated carbon fabric serves as radon collector/radiator, and the SSNTD looks to it. The walls of the compensation module are made of foils of low density polyethylene (LDPE) or high density polyethylene (HDPE). The thickness and surface area of the foil is selected according to the modeling, aiming at achieving the best temperature compensation. The sensitivity of the detector is estimated by its calibration factor CF (CF = net-track density/integrated 222 Rn activity concentration in the ambient air; the net-track density is the number of tracks per unit area, after the background is subtracted). When the detector is outside the detector module, the calibration factor is CF 0 . However, when the detector is placed inside its calibration factor, at equilibrium, will be CF = CF 0 .ρ, where ρ is the "penetration ratio", the ratio at equilibrium of radon concentration in the air inside the module volume to the concentration outside it-in the ambient air. The CF 0 is proportional to the adsorption capacity k and follows the same temperature trend as k, while CF is proportional to kρ. As noted elsewhere [17][18][19] , the fraction ρ that penetrates through the walls in the volume increases with the increase in the temperature-thus the trend with the temperature is reciprocal to that of k. The wall material through which the gas diffuses, its thickness, the wall surface area, and the volume of the "compensation module" are selected so that the temperature variation of kρ is at the possible minimum within the temperature interval of the applications. Modeling. The activated carbon fabrics used are sheets of thickness of few tenths of millimeter, so the process of adsorption may be expected to be fast. Using the model described elsewhere 20 , the evaluation of the time needed the adsorbed radon to reach equilibrium in the activated carbon fabrics was found to be of the order of few minutes, which is negligible compared to typical exposure durations. Further in the mathematical analysis, it is assumed that radon adsorption on the activated carbon fabric sheet occurs immediately, without inertia.
Here, the theory presented in Ref. 19 is modified to consider the adsorption on an activated carbon fabric element of mass m placed in the compensation module volume. A closed volume V (Fig. 1) with a surface area S of the sides through which the radon diffuses is considered. The thickness of the polymer foil of which the sides are made is h, the diffusion coefficient of radon in the polymer material is D. The radon diffusion through the foil is described by the equation: where c is the radon activity concentration in the polymer material and λ is the decay constant of 222 Rn. Equation (1) is considered with the initial condition c(t = 0, x) = 0 and boundary conditions c(t, x = 0) = Kc out (t) and c(t, The quantities c out (t) and c in (t) are the 222 Rn activity concentrations in the air outside and inside the volume, respectively, and K is the partition coefficient of the foil material (the partition coefficient is the dimensionless solubility of the material that is the ratio, on the border, of the concentration in the material to that in the air). Inside the volume the activity is distributed between air and the activated carbon sheet. Assuming fast distribution, without inertia in adsorption process, the total activity inside is 20 : The time dependence of the concentration inside the chamber volume can be modeled by the equation: The first term in the right-side of Eq. (3) describes the flux by Fick's diffusion from the internal surface of the foil, while the second term is for the radioactive decay. In this report integrated mode of measurement is considered so the net track density is proportional to the integrated radon concentration for a given exposure time t exp . The integrated activity concentrations will be denoted by capital symbols, e.g.: The equations and solutions in present case are equivalent to these described and solved elsewhere 19 , when replacing in them V with V′ = V + km.
The expression for ρ, according to the corresponding expression (Eq. (16) in 19 ), will be: where P = KD is the permeability of the polymer material of walls for radon. As the CF is proportional to kρ, the approach of the module design is to choose, for a given m, a polymer material and values of h, V and S of the foil used, so the expression: to be constant, or to vary minimally with the variation of the temperature. Notably, when V << km (which was the case in present designs), the influence of V becomes negligible and the compensation, for a given m and polymer material, remains to depend only on h and S. The temperature dependence of the adsorption capacity can be expressed as 8,21 : where R = 8.31441 J mol −1 K −1 is the molar gas constant, T is the temperature (°C), Q is the heat of adsorption of radon on activated carbon (23 999 J mol −1 , according to the value given in 9 ). At this step, for the purposes of modeling we used for k also data given in the literature, assuming k = 4 m 3 kg −1 at 21 °C (according to 20 , where this value is recommended as a typical value at room temperature). Thus, the corresponding value of the constant κ in Eq. (7) will be κ = 2.19 × 10 -4 m 3 kg −1 . The values of ρ were modeled using data for radon permeability and penetration through LDPE and HDPE, described in 19 and the algorithm, described there. This way η(T) = k(T)ρ(T) was calculated for temperatures within the interval 5-35 °C.
Experimental. Experiments were made at 3 different temperatures (kept constant within ± 0.5 °C): 5 °C, 21 °C and 35 °C using the laboratory exposure facility described elsewhere 22 . The exposures were made at inte- During exposure the activity concentrations of 222 Rn were followed by a reference radon monitor AlphaGUARD PQ 2000 PRO Rn Tn (Genitron GmbH). Pairs of one packed in the module and one non-packed detector with each kind of activated carbon fabrics used, were loaded in the exposure chamber. Another, "background" set of the same pairs was left outside the chamber-at low radon levels (about 20 Bq m −3 ) but at the same temperature as in the exposure chamber. After exposure all exposed sets were left for one week to degas at low radon levels (20 Bq m −3 ) at the same temperatures at which the exposure was made. After out-gazing, the sensors were dismantled and the LR-115/II detectors were removed and etched at 10% NaOH for 100 min at 60.0 ± 0.1 °C. After etching they were washed for 30 min in flushing water and left for 2 min in 50% ethanol. After that they were left to dry and the tracks were counted visually by a microscope to obtain the track-density. The track-density of the "background" detectors, which is due to the detector background and the low radon exposure during detectors storage for out-gasing, was subtracted from the track density of exposed detectors to obtain the "net track-density".

Results
Compensation module design. To verify the assumed temperature trend of k (the temperature trend in CF 0 is the same) the experimental results of detectors exposed outside the modules were used. The results are illustrated in Fig. 2. As seen the experimental results fit well the theoretical curve.
The modelling used to design the individual compensation modules can be illustrated on the example of adsorber of Kynol 507-10 fabric of mass m = 500 mg and a polymer foil of 9 µm thick HDPE foil. As the real compensation modules used in the experiments were wrapped to the sensor so that V < < km, the same was assumed for the considered module (for the assumed mass of 500 mg km = 2000 cm 3 at 21 °C). Criterion for the optimum design was based on the minimum of the relative "standard deviation" (RSD) of η over the temperature range of interest (T 1 , T 2 ), given by the expression: where η is the average value of η = kρ over (T 1 , T 2 ) . In the present research the temperature range of interest was 5-35 °C. In such case, the best compensation, according to the "minimum RSD" criterion was reached when S = 290 cm 2 , e. g. each of the sides of the envelope ("the compensation module") is of area of 145 cm 2 , which corresponds e. g. to a square of side of about 12 cm (more precisely: 12.04 cm). Figure 3 illustrates the results. According to the model the RSD of kρ, and therefore that of CF, over 5-35 °C range may be reduced to 7.2% (the ratio maximum/minimum for kρ in the temperature range specified is 1.22, while that of k is 2.75, i. e. the value of k at 5 °C is 275% that at 35 °C).

Experimental results. For experiments individual modules were designed for detectors using fabrics of
Kynol 507-10, ACC-10 and ACC-20. For Kynol 507-10 and ACC-20 the modules were of HDPE and for ACC-10 of LDPE. The individual design followed the methodology described above, taking into account the exact The results for Kynol 507-10 are illustrated in Fig. 4. The CF 0 at 5 °C is by 305% higher than CF 0 at 35 °C (maximum/minimum = 3.05). In contrast, for the sensors placed in the compensation module the variation of CF is within 17% (max/min = 1.17). Notably, the paired differences between the CFs of the sensors exposed at different temperatures inside the modules were not statistically significant at 95% level of significance.
The results of ACC-10 were obtained with modules of LDPE and are illustrated in Fig. 5. Outside the module, the variation of CF 0 is 265% (max/min = 2.65). In modules, the variation of CF is within 6% (max/min = 1.06), and the paired differences were not statistically significant.
The experiment with ACC-20 was with modules of 9 µm HDPE and the results are illustrated in Fig. 6. In this case for CF 0 the variation is 230% (max/min = 2.3). Again, inside the module the temperature dependence  www.nature.com/scientificreports/ is practically fully compensated, the variation being within 9% (max/min = 1.09) and the paired differences were not statistically significant. The CFs of described detectors with compensation modules are significantly higher than these of other commonly used passive radon detectors. The average CFs with different fabrics for the studied compensation modules are as follows:   www.nature.com/scientificreports/ As seen, detector modules with Kynol 507-10 or ACC-10 provide 7-9 times better sensitivity than that of commonly used passive radon detectors. Looking at the results for detectors exposed outside the modules (Figs. 4, 5, 6) it can be seen that CF 0 at one and the same temperature differs between fabrics. This implies some difference in the adsorption capacity of the different fabrics. One may provisionally rank the adsorption capacity as ACC-10 ≥ Kynol 507-10 > ACC-20. In all modules design the value of the adsorption capacity at 21 °C of 4 m 3 kg −1 was assumed. However as our experimental results as well as the literature data suggest possible variances of k of different activated carbon materials. The activated carbon data in the literature show variations in the range 2-6 m 3 kg −1 for adsorption capacity at room temperature 20 . To clarify the possible influence of the uncertain data for k on the quality of module designs, computer simulations with different k-values were made. The results can be illustrated on the example of the HDPE compensating module modelled in Fig. 3. The results for different k-values are illustrated in Fig. 7. As seen, for all values of k within the interval 2-6 m 3 kg −1 , the variation of η (e. g. the ratio max/min) with the temperature remains much less than the variation of 230-305% observed with the unpacked sensors.
This can be seen clearly in Fig. 8, which illustrates the RSD (η) in the temperature range 5-35 °C, as dependent on k. At k = 4 m 3 kg −1 it is 7.2%, and at variations of k in the indicated interval of k-values RSD does not exceed 13.2%. It can be concluded that the influence of inaccuracies in k-values used for the design of the compensating module on the quality of the temperature compensation is relatively limited.

Discussion and conclusions
A novel highly sensitive passive radon detectors based on activated carbon fabrics with compensation of the temperature influence is proposed. The detectors studied in the present work are made of activated carbon fabric + SSNTD. The detector is placed in a "compensation module" 16 that reduces significantly the temperature influence on the adsorption capacity of the adsorbent used. Modelling revealed that with such design the temperature variation (max/min) of CF within 5-35 °C range may be reduced from 230-305% to 6-17%, according to the experimental results. Such radon detectors may be made flat, very compact and flexible. They demonstrate substantially higher CF than that of other conventional radon detectors 19,23 and this may pave new roads for application. For instance, the detection limit for 6 months exposure, for the radon detectors reported in 23 is 4 Bq m −3 . The sensitivity of other passive radon detectors is comparable to this estimate (e.g. 30 Bq m −3 for one month of exposure, which is 5 Bq m −3 for 6 months as summarized in 1 ). With 7-9 time better sensitivity, the design presented here has potential to achieve detection limit of < 1 Bq m −3 for six months of exposure and few Bq m −3 for one month. The sensitivity might be increased even more by using large area detectors of low background, as those described in 24 . Future research will be focused also on the possibility to use active detectors (e. g. semiconductor detectors of alpha particles) with the compensation modules described.
One more benefit of these modules is that the loss of adsorption capacity due to the humidity is also expected to be greatly retarded, as polymer foils provide efficient, although not absolute, protection against moisture 25 . So far the humidity influence is the factor that limits the exposure duration of activated carbon based detectors to about one week maximum, and even less than one day at high humidity levels 8 .
Nowadays, radon detectors are required to have low interference of thoron on the signal 26 . The novel design provides also efficient protection against thoron ( 220 Rn) interference on the radon signal: Because of its short half-life (55.6 s) thoron almost fully decays during diffusion through the polymer foil of the walls of the module, which is not the case for radon. Using Eq. (5) with the decay constant of thoron, it was calculated that, for the module illustrated in Fig. 3 and for temperatures in the range 5-35 °C, the penetration ratio for thoron is 0.007-0.15%, while for radon it is 30-90% (see Fig. 3).
These qualities make possible the design of a new generation of highly sensitive radon detectors and monitors, applicable for short as well as for long-term measurements under wide range of environmental and temperature conditions. Novel designs of sensitive radon detectors are explored as it is planned to include such monitors in climate monitoring networks, including the European Integrated Carbon Observation System (ICOS). The detectors proposed in the present work may be suitable for that purpose. This however needs extensive dedicated research on the detector background, the possibility to use large-area detectors of very low intrinsic background 24 and the humidity resistance of different compensation modules. Different kinds of activated carbon materials and, possibly, other adsorbents should be studied to find out the most efficient radon adsorbent. It should be noted, that so far the design is based on crude data for k and radon permeability of HDPE and LDPE. More detailed experimental studies of these quantities at different temperatures and for different adsorbents and plastics are planned.
One may speculate on the possibilities to use similar "compensation modules" for detection of the environmental radioxenon (mostly 133 Xe). Radioxenon attracts substantial attention in the light of current challenges in nuclear security and Treaty on the Non-Proliferation of Nuclear Weapons and the Comprehensive Test Ban Treaty 27 . In this case the module should be coupled with a proper beta/gamma radiation detector. Progress in this direction would need dedicated research on the adsorption capacity of different adsorbents for xenon and its transport properties in different polymer materials.
The proposed detectors with compensation modules provide highly sensitive option for monitoring radon and, possibly radioxenon, too in the environment. However, practical application of such "compensation modules" may need some more data on adsorption properties of the prospective adsorbents at different temperatures, as well as on solubility and diffusion of radon and, possibly, other radioactive noble gases too in different polymer materials.